Fluke 190 ScopeMeter Series II RecallPosted on October 26 2012
Dear Fluke Customer:
Fluke is voluntarily recalling its 190 Series ScopeMeter®100MHz and 200MHz 4 Channel Hand Held Oscilloscopes that contain firmware version V10.40. This includes all units manufactured after February 5, 2012 and all units upgraded to firmware version V10.40 after February 5, 2012
Description of the problem
Four channel models with the latest firmware (V10.40) may display the decimal point of the measurement value in the wrong position. This poses a potential safety risk for the user, as an unsafe voltage may be indicated as a much lower (safe) voltage. THIS RISK COULD RESULT IN SERIOUS INJURY OR DEATH.
The affected models numbers are listed in the table below. You can determine whether your 4-channel ScopeMeter® is being recalled by checking the firmware version as described below. Stop using it until after that determination is made.
Model Numbers DescriptionEnding Serial No.
FLUKE-190-104/ SCOPEMETER® 4 CHANNEL 100 MHZ COLOR
FLUKE-190-104/S SCOPEMETER® 4 CHANNEL 100 MHZ COLOR
FLUKE-190-204/ SCOPEMETER® 4 CHANNEL 200 MHZ COLOR
FLUKE-190-204/S SCOPEMETER® 4 CHANNEL 200 MHZ COLOR
This recall does not include the 2 channel ScopeMeter® units.
Identifying the recalled product
To determine whether your ScopeMeter® is being recalled, first check the firmware version of the product by following these steps: (1) Disconnect the ScopeMeter® from any signal source; (2) with the unit turned on and facing you, press the user key located at the bottom right side of the instrument's front panel; (3) press F3 (Version and Cal).
The firmware version is then displayed as shown below with other instrument information
FIRMWARE VERSION DISPLAY
Stop using the Fluke 4 channel ScopeMeter® even if you have not experienced problems.
Verify your 4 channel ScopeMeter® is affected by this recall using the steps above.
Either return your 4 channel ScopeMeter® for a free upgrade or go to 190fix and follow the instructions to download a new version of the firmware
You can return your ScopeMeter® by contacting your local service center or calling 61 (28) 850-3333, 9 a.m. to 5 p.m. AEST Time Monday–Friday.
We are undertaking this voluntary recall because we place tremendous importance on our customer's safety and the reputation of our products. Please accept our apologies for any inconvenience caused by this action.
James L. Cavoretto
Sr. Vice President
Chief Technical Officer